I'm completely caught up on UMIK-1 orders (and almost caught up on EMM-6/Galaxy orders, I should get through my backlog by tomorrow). If you ordered one but haven't received an email from USPS with your tracking information, PM or email me and I'll send your tracking number.Did you receive an email for shipping and how many days did it take from the time you ordered? I just ask because it was said that supply was limited and wondering if I will get mine any time soon.
According to miniDSP, newer mics have a default attenuation of -12 dB (that should show up when you select the microphone in the input device settings of REW or whatever program). If you are getting a max SPL of 105 dB, 105 dB + 8.4 dB (factory sensitivity) + 12 dB (default attenuation) gives 125.4 dB, which is pretty close to the 130 dB spec.Got mine today from CSL! Did a couple measurements. The sensitivity thing is throwing me off a little. Factory cal file says -8.4 dB. CSL cal file does not include. Input signal is well below the output signal when setting levels. SPL cal was done with a RS meter and afterwards says max SPL is 105 dB. Mic is supposed to be good to 130 dB so I'm thinking I need to add the factory sensitivity to the cal file?
Yes, but it seems to limit my measurement capabilities (SPL). I think I'm going to try and add the factory sensitivity cal to the CSL cal file. I can still verify with my RS meter.Did you receive an email for shipping and how many days did it take from the time you ordered? I just ask because it was said that supply was limited and wondering if I will get mine any time soon.
Also, how about some screen shots of the calibration files to compare against.
I understand that for now the Sensitivity data will not accompany the CSL cal. file due to issues with potential conflicts of other accoustic analysis software and some other details. So, SPL should just be calibrated as you did with your RS meter against REW's SPL software meter
Copy and paste what ever is in the Minidsp file into the top of the CSL file (first line).Was wondering...In order to add the MiniDSP sensitivity number to the CSL cal file, do I just copy-n-paste the "Sens Factor =-#dB" line to the top? Do I need the quotes? And will it throw anything off?
Thanks, I suspected that would do it. Just wanted to be sure. :sn:Copy and paste what ever is in the Minidsp file into the top of the CSL file (first line).
All it does is adjust the SPL, to be close to absolute.
If you have a SPL meter you can double check.
Thanks, I suspected that would do it. Just wanted to be sure. :sn:
BTW, do you guys use the little foam windscreen that came with the mic?
I don't for my other mic.
I would email or post Herb, depends if he used it when he calibrated it. I would doubt that he would off.
Thank you for the info, good to know that.I don't use the windscreen when making the measurements. It basically makes a roughly 0.5 to 1 dB difference above 15 kHz or so:
For vocal mics (which the UMM-6/UMIK-1/EMM-6/ECM8000 are definitely not) true, although again, the one that comes with these low-cost measurement mics are so small I doubt they would provide any benefit.I kinda thought it was also to reduce breathing noises from humans that are speaking/singing directly into the mic... which is also turbulence related.
Protection is the only reason. The mic is certainly calibrated without windscreen in place. With a calibration mic, breath noise is not an issue, and wind noise is not an issue unless making speaker measurements outside to escape room effects.The purpose of windscreen, other than protection?
AgreeProtection is the only reason. The mic is certainly calibrated without windscreen in place. With a calibration mic, breath noise is not an issue, and wind noise is not an issue unless making speaker measurements outside to escape room effects.
I would leave it on, though, for a little extra protection from dust and gunk, and from me accidentally bumping or dropping it on a klutzy day. The minor rolloff above 10 KHz is annoying but of little actual consequence. My 2 cents worth.
Depends what batch you have.I am really disappointed I ordered mine from miniDSP and not CSL ( was before they sold them). After my order was placed I contacted CSL and found out they were going to provide calibrated UMIK-1, what a bummer. It's not cost effective to send mine in for a calibration at $55. :no: